Message 4 of 9. Do not use the sequence structure for what you’re doing. Well, I repeated those exact same steps with that driver from that location and there is no dll included in the zip and dropping any function on the block diagram prompts a search for the dll. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 3 of 4.
|Date Added:||6 August 2014|
|File Size:||61.65 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
If the Initialize function does the same thing, then’s its possible you need to configure the instrument’s GPIB parameters. The thing I am working on right now is to restart the phone if it failed oabview the testing. What Dennis says may also be a reason for the age one.
The link on NI’s page to get the DLL doesn’t seem to be working but you should be able to go to Agilent’s web site www. This term character may be added by the higher level functions in the driver but this call is pretty low level and you’re expected to know the exact syntax when using it.
Message 3 of 4. The only way to currently restore manual control is to completly disconnect the GPIB cable, or close every single NI application open stopping the labview program is useless – labview in its entirety, even the welcome window, must be closed before control is restored. It’s possible that the instrument requires a special termination character at the end of each command.
Most Active Software Boards: Message 2 of 5. I have three frames in a flat sequence structure, the first two are for start up with parameters I needthe third one contains a while loop, inside which is my testing code, I am going to put my restart code there. But, when I try to work with them it asks for some more VIs.
Well, I repeated those exact same steps with that driver from that location and there is no dll included in the zip and dropping any function on the block diagram prompts a search for the dll.
I then went to the agilent. What you need to do is just use a Labview Interface Generator to convert the. This prototype was completed in mid, and has been used by a few customers. I installed this one.
Automation for Agilent EC – Discussion Forums – National Instruments
Any idea is well appreciated! Message 4 of 4. I cannot find the driver of There’s also the JKI state machine. Also, this is an excellent resource for learning about state machines. Hi, I am writting a program in Labview 8. Your best option is to go with a state machine.
Message 4 of 4. Thanks for your reply. Did you ever get a driver? But I think the and together can realize the function of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Agilent Technologies / Keysight Technologies E1968A
Message 2 of 4. Accepted by topic author DennonO. I already put in a request on the NI driver request page.
Most Active Hardware Boards: Message 3 of 4. What could be the issue?